Mileage measurement Remove the test strip from the vial , immediately close it . Hold the test strip so that the orange field test was directed upwards. Carefully insert the test strip into the guide bar in the direction of the arrow until it clicks. This means that the test strip was in the right place . Inserting the strap to slide automatically activates the camera . After a short display test and check the code number ( should correspond to the code on the vial of test ) , the display shows the symbol of the test strip and flashing blood drop . Apply a drop of blood on the middle of the orange box , and your finger . The result you get after 5 sekundach.Aparat automatically turn off after removing a test strip . The result of the measurement date and time will be automatically stored in memory . Application of blood outside the measuring apparatus Remove the test strip from the camera is displayed when the test strip and flashing blood drop . Apply blood to the test strip and insert it back to the camera in about 20 seconds .
Visually check the reliability of the test result Check the color of the circular viewing window located on the back of the test strip . Compare the color of the window with the color scale on the label of the container of test strips to check the reliability of the result. Note : Decisions about treatment should be made only on the basis of the results displayed by the measuring apparatus . For more information on how to use the measuring apparatus can be found in the instruction manual Accu-Chek acive .
Specifications: Measuring principle : determining the level of glucose in fresh capillary blood by reflectance measuring range: 10-600 mg / dl or 0,6-33,3 mmol / l Blood sample size : 1-2 microliters measuring time: about 5 seconds . Power supply: 1 battery ( type CR 2032 ) Battery capacity : 1000 measurements or 1 year Automatic shutdown: after 30 or 90 seconds . according to the most recent operation Memory: 350 results with the date and time, and the average of 7 , 14 and 30 days Temperature: range: from 10 ° C to + 40 ° C , during storage without batteries : -25 ° C to +70 ° C , during storage of the battery , from -10 ° C to +50 ° C humidity: during the measurement up to 85 % relative humidity, and during storage up to 93 % relative humidity Height : up to 4000 meters above sea level Dimensions: 104.5 x 51.5 x 22 mm weight: without batteries approximately 55 g with battery 60 g display : 96 -segment liquid crystal display LCD screen with icons transfer data to the PC using the infrared port